||Purpose of Test
||Data used to maintain alkalinity at a fixed level to minimize corrosion in distribution systems (treatment strategy to help comply with the Lead/Copper Rule). The data is also required to calculate the % TOC that must be removed under the Enhanced Coagulation Requirement (D/DBP Rule Stage 2) for surface water facilities.
|Data used to control Chlorine based disinfection processes, required for reporting at POE and in the distribution system, also used in monitoring in the distribution system during flushing and line break maintenance. EPA Primary DW Standard at 4.0 mg/L as CL2.
||Data used to control Chlorine based disinfection processes, required for reporting at POE and in the distribution system, also used in monitoring in the distribution system during flushing and line break maintenance. EPA Primary DW Standard at 4.0 mg/L as CL2.
||Data used to monitor particulates / microbes; Per the Interim Enhanced Surface Water Treatment Rule — Turbidity measurement is required for Conventional and direct filtration systems on each filter and on the combined filter effluent. Also used to monitor raw water changes, signal storm events and to support watershed protection plans.
||Turbidimeter (2100xx) See instrument manual
|Data used for monitoring of pH throughout the plant, controlling processes and corrosion.
||pH meter & electrode - see instrument manual
||Data used to control concentration of fluoride for dental protection, to monitor for high natural levels in raw water and to optimize fluoride feed process. Primary DW Standard MCL at 4.0 mg/L Fluoride.
||Data used to monior residual coagulant (alum (aluminum sulfate) is most commonly-used coagulant). Secondary DW Standard regulation at 0.05 to 0.2 mg/L as Al.
||Data used to monitor manganese for stain control. At 0.05 mg/L, it causes black stains on plumbing fixtures, laundry and other items in contact with the water. Secondary DW Standard at 0.05 mg/L Mn.
||Data used to monitor iron in ground water systems, iron carry-over from iron-based coagulants, iron in distribution systems as a corrosion by-product from cast-iron or steel pipes. Iron is also a food source for iron bacteria which form large masses that clog well screens, pumps and other equipment. BART method for determining iron-reducing bacteria available. Secondary DW Standard at 0.3 mg/L Fe.
|Chloramine, Free Ammonia
||Data used to optimize formation of monochloramine and minimize level of free ammonia during chloramination process and monitor free ammonia levels in the distribution system to determine the potential for nitrification (applicable to surface water facilities and booster stations of chloraminated water systems).
||Data used to monitor for watershed protection, source water (especially in agricultural areas) following storm events and spring run-off conditions and potential for methemoglobinemia (blue baby) condition. EPA MCL at 10 mg/L Nitrate Nitrogen.
|Calcium (Ca Hardness)
||Data used to monitor the effectiveness of hardness removal or scaling control treatment practices. Hardness is also monitored to check effectiveness/ regeneration of ion-exhange processes (water softeners).
||Data used to monitor and control removal of color from raw water, typically by chemical gagulants or pre-oxidants. Secondary DW Standard at 15 Platinum-Cobalt Color Units.
|Chlorine, Chlorine Dioxide Amp Titration
||Data used to control disinfection processes when used as preoxidant to reduce DBP formation. Chlorine Dioxide is also used in manganese control. Chlorite is a by-product of chlorine dioxide and must also be checked daily under D/DBP Rule. Chlorite MCL is 1.0 mg/L as chlorite.
||See AutoCAT 9000 manual.